This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||4 June 2015|
|PDF File Size:||9.43 Mb|
|ePub File Size:||1.7 Mb|
|Price:||Free* [*Free Regsitration Required]|
For example, when you press the key on the menuit will bring up its own menu of Orientation To print the manual completely, please, download it.
Used Agilent EA
Used on Option Noise Figure personality. It is effective over a frequency range of dc to 2 GHz. Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer. Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger maanual Trace Average Display line annotation Turning on the Analyzer for the First Time Plug in the power cord.
To access the menu, press Rename 2.
Select the type of file you want to load setup, state, trace, limits, or corrections. Preselection reduces mixer overload from broadband signals, reduces radiation of local oscillator harmonics back to the device under test, and reduces the level of image and multiple responses manuxl.
Set the frequency span to 50 MHz: To access the Load menu, press 3. Page 30 Front and Rear Panel Features Key Overview In other key menus, one key e44440a will always be highlighted to show which key has been selected but the menu is immediately exited when a selection is made.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
Notice The information contained in this document is subject to change without notice. Page 18 When you have completed msnual printer setup for the analyzer, press the front panel key. Page 41 Improving Frequency Accuracy If the printer is ready and the printer setup was successful, a printout of the mahual display will be printed.
Press If the previously performed alignments did not resolve the problem, press Alignments Restore Align Defaults Is the analyzer displaying an error message?
You can order the service documentation for Agilent spectrum analyzers through your Agilent Sales and Service office. Reset the analyzer by pressing 2.
Page of 80 Go. Viewing A Signal Viewing a Signal 1. The only green key is the key, which performs an analyzer reset A summary of all front Preset Is the analyzer turned on? Press System Reference press the softkey to toggle the reference on. Enter text from picture: Increments or decrements the e4440s value. The front-panel keys are dark gray, light gray, green, or white in color.
In this example you are loading a trace file; This analyzer does not contain customer serviceable fuses. Arrow Keys Numeric Keypad Enters a specific value. Save the file 1. Agilent handheld spectrum analyzer user’s guide pages. Got it, continue to print. The following table describes the four types of spectrum analyzer messages.
Installation and Setup Turning on the Analyzer for the First Time If using LAN, set the IP address of the analyzer to an appropriate number for your network one that the network recognizes, but that is not yet in use: File Menu Functions This chapter describes how to use the functions located under e4440x front-panel Data storage and retrieval are handled similarly to that of personal computers PCs.
RS This feature not fully implemented; to be implemented in a future release. In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem.
Please be as specific as possible about the problem.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
Front and Rear Panel Features This chapter gives you an overview of the front and rear panels of your analyzer. CAUTION The analyzer is mankal with a a transportation disk inserted in the disk drive to prevent damage to the disk drive during transportation. Restores the analyzer to its factory-defined state. Like PCs, these analyzers include an internal storage drive and a floppy disk drive, both of which have directory and sub-directory capability.
Installation and Setup Running Internal Alignments Running Internal Alignments Each time the analyzer is powered on, the internal alignment routine runs automatically. Page 70 In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem.
This symbol is used to indicate the ON position of the power button. In Case of Difficulty Returning an Kanual for Service Returning an Analyzer for Service Service Tag When you are returning an analyzer to Agilent Technologies for service, fill manua and attach one of the blue service tags provided at the end of this chapter.
EA 3 Hz – Refer to Figure Options and Accessories Ordering Options and Accessories Ordering Options and Accessories Options and accessories help you configure the analyzer for your specific applications, and enable you to use the analyzer. Each softkey More has several characters on it.